Laser mold equipment, separation refrigeration system, optical path system, control system and energy system.
Cooling system : to guarantee the regular replacement of cooling water ( or pure water distillation ), regular replacement of filter. In normal working three hours, refrigeration machine actual temperature and set temperature difference does not exceed 10℃as normal.
Optical path system: optical path from the back water pipe interface to the microscope, followed by a red light indicator, back reflecting mirror, small bracket, cavity, before the mirror, a beam expander, 45 ℃mirror, a focusing mirror.
Should guarantee the lens without the attachment of dust, no scaling, no dilapidation, xenon lamp should be replaced regularly,regular paper with light detecting optical, laser in photometric paper spots smooth, uniform, no gap is normal.
Control system: in the process of operation, to protect people from the laser beam light injury, protect the welding point is not oxidized, the welded surface appearance, smooth, visual processing material, adjusting the energy of the laser beam size, ensure the welding effect
Rate. Should ensure that the display data clear, all the parameters of flexible adjustment, valve normal work, filter working normalfoot from the normal, light, and laser beam bursts of points evenly, there is no obvious difference between the spot energy.
The current system is to provide: a laser beam energy system preparation time is about 60 seconds, start, start button indicating lamp should always be bright, after the success of its normal combustion, alarm buzzer, regulating pulse width or charging voltage
Current, should be " drop " in response to the acoustic. The discharge process in xenon lamp ends should be blue, shall ensure that each laser beam energy evenly, and according to the control system to regulate the different parameters of normal variation.
Working table for processing parts: operation platform.
Ensure that the X axis and Y axis, electric and manual operation, long falls, flowing freely. If the system does not meet these requirements, may refer to the following solution for.